Electromagnetic flow converter for combination with OPTIFLUX and WATERFLUX flow sensors. KROHNE 03/ – – TD IFC R05 en. The documentation is only complete when used in combination with the relevant documentation for the . 2. The high-performance solution. IFC is the only electromagnetic flow converter with 3 x diagnostics of instrument, application and .
|Published (Last):||8 September 2016|
|PDF File Size:||19.83 Mb|
|ePub File Size:||8.81 Mb|
|Price:||Free* [*Free Regsitration Required]|
Current not correct, mA output cable has open circuit or load too high.
Construction of signal cable B 1 Stranded drain wire for the inner shield 101. About Us Vissers Sales Corp.
IFC Signal Converter From KROHNE
Automatic range conversion via threshold or control input. Electrical connection General Electrical connection is carried out in conformity with the VDE directive “Regulations for electrical power installations with line voltages up to V” or equivalent national regulations.
A shielded two-wire copper cable is used as the field current cable. Using An Ir Interface option For information about possible status messages, their meaning and cause refer to Status messages and diagnostic information on page Using a suitable tool, pull out the two metal puller devices to the left and right of the display.
Krohne IFC Signal Converter for Electromagnetic Flowmeters – SRP Control Systems Ltd.
Note the key function within and between the columns. Electrical krhone is carried out in conformity with the VDE directive “Regulations for electrical power installations with line voltages up to V” or equivalent national regulations.
An external power supply is necessary to operate activation the subsequent devices Uext. S pipe not full Only for measuring sensors with 3 or 4 electrodes. Preparing the interface cable Figure There is an additional bush for the interface cable.
Its measurement performance is excellent even in more difficult applications like mediums with low conductivity or mediums with high solid content or entrained air, corrosive and or abrasive mediums. It may be necessary to activate resetting of the counter in the menu “quick setup”.
Changing of settings in Fct. In terms of available housing versions, there is a compact variant, in which the signal converter is connected to the ofc sensor, as well as the field, wall-mounted and 19″ rack-mounted housing. Page 34 C 1.
The signal cable does not iff any halogens and is unplasticized, and remains flexible at low temperatures. For further information on electrical connection refer to Description of the inputs and outputs on page Shown is the BTS cable.
This warning refers to the immediate danger when using this device in a hazardous atmosphere. If you nevertheless use shielded cables, the shield must NOT be connected in the housing of the signal converter.
Counters did not run during it.
KROHNE Ifc-300 F ALTOMETER Electromagnetic Flowmeter Signal Converter B437790
Only available when linearity test “on” activated in Fct. The grounding cable should not transmit any interference voltages. Or electrodes completely insulated e. Tag Ifd point identifier Tag no. The operating point of the 4 optical keys is located directly behind the glass pane. The electrical connection of the outer shield is different for the various housing variants. Shield connected using 6.
Remove any shield that is present. Clamping of shields 1 Field current cable 2 Signal cable At flow sensor side: The manufacturer can not accept responsibility for any damage or injury caused by misunderstanding of the information in this document.