transformer, test leads, and input jacks — must be purchased from Keithley Instruments. Standard fuses with For analog filter (Models A/A only) . leaving the factory the A and A meet these specifications. the Series A Reference Manual for additional power derating. Keithley A Pdf User Manuals. View online or download Keithley A User Manual.
|Published (Last):||24 November 2010|
|PDF File Size:||13.81 Mb|
|ePub File Size:||14.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Keithley 2636a reference manual
After the specified warm-up period. TSP command reference Example 2 printbuffer 1, 10, smua.
Mitochondrial dna polymerase is heterotrimeric, consisting of a homodimer of accessory subunits plus a catalytic subunit. Instead, access the global variable directly. In the IP address field, enter Keithley a series reference manual pdf download. TSP command reference os.
All delay values are specified in seconds. Simultaneous control of multiple instruments over visa. Compute the current reading from the DMM voltage reading and characterized 0. When a static trigger configuration is not possible, you can use the interactive triggering method to control the timing and actions of the SMU.
D-1 Script command equivalents Turn On Power Select and set the source level Perform the following steps to select the source and edit the source value: It does not wait for command completion on the remote instrument. Configure recommended instrument settings before making connections to the device.
Set bit indicates that the SMU B current limit was exceeded. TSP command reference In addition to the above constants, operationRegister can be set to the numeric equivalent of the bit to set.
Using labview to program kethley a to produce voltage. For connection information, refer to Contact check connections on page The min and max entries each have the attributes defined in the following table. In addition to the above constants, questionableRegister can be set to the numeric equivalent of the bit to set.
What’s missing? Tell us about it.
It is typically set to the date when the next calibration should be performed. Keithley picoammeter manual, brochure and application notes.
Functions and features When the smuX. Iv sweeps using the keithley and labview start up. Y on page bufferVar. Functions and features SMU event detectors As shown in the Using the remote trigger model on page topic, the source-measure unit SMU has multiple event detectors see the table below in order to control the timing of various actions. Section 2 General operation In this section: Select Mznual dialog box 3.
The following programming example illustrates how to request a Channel A current reading: Configure the timer to start when the sweep is initiated. Commands that cannot be used in scripts General commands abort endflash endscript flash loadscript loadandrunscript password restoreglobals IEEE Std We take purchase orders and can provide special school, government and corporate pricing.
Mwnual operation Operation overview Before you begin any of the following front panel procedures, make sure that you exit out of the menu structure. Empty the buffer using the bufferVar.
Be sure to use appropriate shielding, guarding, and aperture selections when making low-current measurements. Reading the system node enable register returns a value.
KEITHLEY B : User manual
What Are The Source-measure Capabilities Hall effect measurements for semiconductor and other material characterization robert green keithley instruments, inc. For these examples, the System SourceMeter instrument is programmed to source mA and limit voltage 10 V.
Search for your model number’s firmware: Calibration Voltage source accuracy Follow the steps below to verify that the Series B output voltage accuracy is within specified limits. General operation Requesting readings You can request readings by including the appropriate measurement command as the argument for the print command.
You can also add or remove a node to or from a group at any time.
Additional Series B information Provides unpacking instructions, describes basic connections, and reviews basic operation information. When proper connections are made, the instrument chassis is connected to power-line ground through the ground wire in the power cord. Send the commands that need to be included in the script. Functions and features Understanding hardware value and pseudo line state LAN triggering is similar to hardware synchronization except that LXI trigger packets are used instead of hardware signals.
It runs after all the scripts have loaded and any scripts marked as autorun have run. TSP command reference lan. The following programming example illustrates how to display the last key pressed: