Flir Systems has launched the SC thermal-imaging camera, which is designed to deliver a combination of infrared and visible spectral images. The portable. Its FireWire interface can transfer bit radiometric or RAW data directly into a PC for real-time analysis of captured images and for FLIR SC/SC also. Buy SC 45°, Thermal Imaging Cameras, 30 Hz Image Frequency, 45° FOV or other Radiometric cameras – uncooled products from Acal BFi BE. Get a quote.
|Published (Last):||3 September 2013|
|PDF File Size:||6.78 Mb|
|ePub File Size:||16.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
If you keep the camera in a warm room, it will overestimate temperature by an average of 1.
Last summer, I had a high school volunteer Padraic Odesse join my lab for a few hours a week to help out my graduate students with data analysis. I am not affiliated with FLIR, nor do I receive revenue, salary, or funding from them, so this post should not be interpreted as an endorsement.
Depending on your measurement purposes, flri might be ok, but the best error you will get under optimal conditions appears to be greater 0. On the fli, both cameras perform rather well under these optimized conditions, although clearly the SC performs better. Comments Be the first to write your review!
The mean absolute error will tell you whether the camera is reading higher or lower than actual temperature absolute value simple removes the negative sign. FLIR launches firefighting training centres-schools program.
Strangely it does not seem to converge toward the real temperature within our 10 minute long experiment. Because they are adjacent to each other, they very nearly capture a similar field of view.
Contact supplier for specifications here. Certainly not as inaccurate as I had feared it might be except see below. But, remember, we were performing the analysis sc6660 thermography software that allows for some correction of environmental variables, and we were constantly tuning the FLIR ONE.
If the reference signal is at a known and constant temperature, you can, in essence, calibrate the thermal detector to a more stable source. So, my lab assistant diligently captured an image with the two cameras, taken at the same distance and at 8 different water temperatures 5 to 40C.
Email required Address never made public. The one lens is the lens that allows long wavelength infrared radiation to pass through to the detector and the other lens is probably very similar to an iPhone camera.
Sc66 is usual with cell phones, people are likely to keep the device in their pocket, or maybe leave it exposed to the elements. The attachment is actually two cameras in one. The PiP function has full flexibility allowing users to adjust the PiP area size and position. New and innovative range of FLIR thermal imaging cameras now available.
I would not trust the spot temperature measurements. Be the first to write your review!
Flir SC660 thermal-imaging camera is suitable for demanding R&D applications
Large LCD screen Super size 5. Picture-in-picture Create an infrared overlay on your visual image. You are commenting using your WordPress. Stay up to date with the latest products like this No, thanks Nice one! Each time it jumps in temperature corresponds to when we tuned it. Look vlir those R2 values!
High end thermal image security cameras now affordable for the home.
Thermal imaging camera / infrared / with LCD screen / handheld – FLIR SC – FLIR SYSTEMS
The digital Firewire interface connects to the PC running ThermaCAM Researcher — powerful thermographic software for real-time flid recording and analysis.
This product is not sold individually. Business registration, contact details and locations have been verified. Save to existing comparison list Select. So we took as realistic an approach as possible ran the measurements 3 different ways. FLIR thermal imaging cameras in automotive quality control. Automatic- and Manual focus, Digital zoom Focus possibilities include; single shot auto focus, continuous auto focus, laser based models or manual focus.
Laser Pointer Helps you associate the hot or cold spot in the IR image with the real physical target in the field. The high quality lenses including USM technology have fast autofocus functions, as well as full manual adjustment. If you narrow in on how fliir the sc60 is this will make a bit more sense. They are the perfect choice when a portable infrared camera is preferred.
FLIR thermal imagers can cut power bills. The reason things look really accurate near room temperature is because the formula for calculating temperature predicts that if all temperatures are similar to each other, the contaminating radiation signal will be similar to the object signal. But these are engineered devices.
Of course, I would be most grateful if FLIR did give me a free thermal camera or a news lens for all the fine press I give them!
What other thermal imagers tend to do is use a shutter like this as a reference signal, and also for Non-Uniformity Flie of the fir array. The buttons and joystick to control the camera are integrated in this handle and always stay right underneath your fingertips. Visible spectrum images generated by the built-in high resolution cs660 camera can be fused with the IR images creating a Picture-in-Picture PiP view in real time. If you keep the camera in a cold room, it will severely underestimate temperature by an average of 5.
Programmable direct access buttons For increased flexibility the operator can program buttons located on the top of the camera for direct access to favourite functions. Furthermore, radiometric sequences can be stored on high capacity SD-cards.
The FLIR software lets you do neat things with overlaying the images, but to be honest, I find the whole outline overlay to be annoying.
So, if you have other FLIR software, you can extract the actual thermal image and do proper thermography on the image.