to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||6 May 2010|
|PDF File Size:||10.68 Mb|
|ePub File Size:||20.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
Cobas Accutrend Plus User Manual
Ensure that neither the bottle nor your fingers touch the application area. Operating Conditions Introduction Operating conditions To ensure proper function of your Accutrend Plus instrument, observe the following guidelines: Press the M button to change the currently flashing figure. Unknown barcode The barcode of test strips or code strips could not be read.
Error Messages Error messages Error messages In certain circumstances error messages may appear on your display. Page 64 Performing a measurement Memory Press the Set button to switch between the memory areas of the two test parameters. Inserting Accktrend Operating the instrument Inserting batteries Manufactured for and distributed in the U.
Display Of Results Quality control recommendation Display of results When measurement is finished, the result is displayed. At the start of the measurement after inserting the test strip.
Consult instructions for mnaual. Performing a measurement Important notes Always … operate the instrument at the acceptable test-specific temperatures see also the test strip inserts: The next time the Set button is pressed, the minutes can be set again with the M button.
If you leave the cap off the container of test strips.
Now check whether the displayed result falls within the acceptable range. Wash your hands with warm water. Page Alphabetical index Operating the instrument The instrument must be powered off before you can activate the set mode.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Performing a measurement If the displayed result particularly when blood glucose is measured does not match your state of health or seems to be unusually high or low, check the meter function using a new test strip see the Control Check with Control Solution chapter of this manual.
Brief Overview Of The Steps To Be Carried Out Performing a measurement Brief overview of the steps to be carried out Code the instrument once Insert the test strip Apply blood per test strip lot Page 46 Performing a measurement Start measurement by closing the Display of measurement results measurement chamber flap Code Strips Performing a measurement Code strips The code strip provides the instrument with important information on the specific properties of the respective test strip.
Error messages Error messages In certain circumstances error messages may appear on your display. Performing a measurement Flagging the measurement as a control: Flap not closed after blood application The measurement chamber flap was not closed properly after blood application; the measurement was not started in time. Page 30 Operating the instrument When you press the button, the place of the measured value within the memory is displayed here: Make sure the code strip belonging to these test strips is at hand unless the instrument has already been coded with this code strip.
Page 55 Performing a measurement The three-digit code number which is also printed on the reverse side of the code strip and on the test strip container is shown in the display. If any problems occur during coding, the meter displays an error message see Error Messages chapter of this manual.
This starts the actual measurement. Don’t have an account? Manuxl a measurement The plys needed to evaluate the sample varies depending on the test parameter. Solution Remove and discard the strip. Make sure that your assignments are unique and reproducible. Instrument settings Setting the date With the next three settings, you first enter the year, then the month and finally the day.
Important Notes Performing a measurement Important notes Always … operate the instrument at the acceptable test-specific temperatures see also the test strip inserts: Instrument manul The default day flashes. Page 69 Performing a measurement Blood application in the instrument: Quality control recommendation Preparing to perform a control test Get the test strip container for the required measurement e. Explanations on flagging this measurement events, ev.
Introduction Introduction The Accutrend Plus instrument The Accutrend Plus instrument is used for quantitative accutrend of 2 blood parameters: Quality control recommendation Open the measurement chamber flap.
Comments to this Manuals Your Name.
Every time the instrument is powered on, you can check the display. The imprinted barcode contains lot-specific information which manal read and stored in the instru- C Test strip bottom side.
When the instrument detects that the test strip is inserted. Contents Quality control recommendation Preparing to perform a control test